|Publisher DOI:||10.1016/j.arth.2017.05.009||Title:||The Influence of Contamination and Cleaning on the Strength of Modular Head Taper Fixation in Total Hip Arthroplasty||Language:||English||Authors:||Krull, Annika
|Issue Date:||Oct-2017||Source:||Journal of Arthroplasty 10 (32): 3200-3205 (2017-10)||Journal or Series Name:||The journal of arthroplasty||Abstract (english):||Background Intraoperative interface contamination of modular head-stem taper junctions of hip implants can lead to poor fixation strength, causing fretting and crevice corrosion or even stem taper fracture. Careful cleaning before assembly should help to reduce these problems. The purpose of this study was to determine the effect of cleaning (with and without drying) contaminated taper interfaces on the taper fixation strength. Methods Metal or ceramic heads were impacted onto titanium alloy stem tapers with cleaned or contaminated (fat or saline solution) interfaces. The same procedure was performed after cleaning and drying the contaminated interfaces. Pull-off force was used to determine the influence of contamination and cleaning on the taper strength. Results Pull-off forces after contamination with fat were significantly lower than those for uncontaminated interfaces for both head materials. Pull-off forces after application of saline solution were not significantly different from those for uncontaminated tapers. However, a large variation in taper strength was observed, pull-off forces for cleaned and dried tapers were similar to those for uncontaminated tapers for both head materials. Conclusion Intraoperative contamination of taper interfaces may be difficult to detect but has a major influence on taper fixation strength. Cleaning of the stem taper with saline solution and drying with gauze directly before assembly allows the taper strength of the pristine components to be achieved. Not drying the taper results in a large variation in pull-off forces, emphasizing that drying is essential for sufficient and reproducible fixation strength.||URI:||http://hdl.handle.net/11420/2765||ISSN:||0883-5403||Institute:||Biomechanik M-3||Type:||(wissenschaftlicher) Artikel|
|Appears in Collections:||Publications without fulltext|
Show full item record
checked on Sep 27, 2020
Add Files to Item
Note about this record
Items in TORE are protected by copyright, with all rights reserved, unless otherwise indicated.