|Publisher DOI:||10.3389/fsurg.2017.00078||Title:||Assessing the GRIP of Ventral Hernia Repair: How to Securely Fasten DIS Classified Meshes||Language:||English||Authors:||Kallinowski, Friedrich
Silva, T. G.
|Keywords:||GRIP;bridging;fixation;glue;hernia meshes;model for ventral hernia repair;overlap;ventral hernia repair||Issue Date:||19-Jan-2018||Source:||Frontiers in surgery (4): 78 (2018-01-19)||Journal:||Frontiers in Surgery||Abstract (english):||
Recurrences are frequently observed after ventral hernia repair. Based on clinical data, the mesh-defect area ratio (MDAR) can lead to lower recurrence rates. Using dynamic intermittent strain (DIS) in a pig tissue model, MDAR can be modified to give a measure called grip to better assess the mechanical stability of ventral hernia repair. The focus of this experimental study is to assess the different aspects of mesh overlap (OL) and fixation only in bridging repair of ventral hernias. DIS mimics coughing actions in an ex vivo model with the repetition of submaximal impacts delivered via a hydraulically driven plastic containment. Tissue derived from pig bellies simulates a ventral hernia with varying defect sizes. MDAR is calculated from the hernia orifice and the mesh OL. Commercially available meshes were strengthened with glue, tacks, and sutures to bridge the defects. The reconstructions are strained with up to 425 dynamic impacts. The grip of each repair is assessed using MDAR modified by the strength of the fixation. The DIS classification is based on bridging of a 5 cm ventral hernia orifice with an OL of 5 cm in a sublay position. The classification discriminates meshes properties upon DIS strain. MDAR is calculated to be 9 under these conditions. Decreasing the OL or increasing the hernia orifice reduces MDAR to numbers below 9. MDAR is modified to reach GRIP. Closure of the peritoneum adds about 4 to the grip given by MDAR. The multiplying factor of a transmural suture or one tack of Securestrap® or Protack® is 0.5 times the number of tacks applied. The multiplier given by a bonding spot of Glubran® is similar to that of an Absorbatack® being 0.33. Plotting the likelihood of a bridging repair to survive more than 400 DIS impacts versus the grip estimated from the factors given above, the grip to be passed for a durable repair is 10 for Parietex Progrip®, and Dynamesh Cicat® and 25 for Dynamesh IPOM®. Clinical data previously published can be reculculated to assess MDAR and permit an estimation of the grip of the reconstruction. In these recalculations, a correlation between MDAR and long-term recurrence rates is found. A dimensionless number called grip can be calculated. The grip can be modified by fixation in a reproducible way. A higher grip can improve the durability of ventral hernia repair. We believe that a higher grip leads to lower recurrence rates in the clinical setting.
|URI:||http://hdl.handle.net/11420/2935||DOI:||10.15480/882.2343||ISSN:||2296-875X||Institute:||Biomechanik M-3||Document Type:||Article||License:||In Copyright|
|Appears in Collections:||Publications with fulltext|
Show full item record
Files in This Item:
checked on Jan 19, 2022
checked on Jan 19, 2022
checked on Jan 12, 2022
Note about this record
Cite this record
Items in TORE are protected by copyright, with all rights reserved, unless otherwise indicated.